KiWAVE™ is a trademark of Keithley Instruments, Inc. All Keithley Instruments product names are .. Model Arbitrary Waveform Generator User’s Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 The exceptional signal quality of the Model is a result of its high resolution, . Refer to the Keithley Model Arbitrary Waveform Generator User’s Manual if it is not clear how to specify the waveforms. The figure below shows an.
|Published (Last):||3 May 2010|
|PDF File Size:||19.94 Mb|
|ePub File Size:||14.68 Mb|
|Price:||Free* [*Free Regsitration Required]|
Comments to this Manuals Your Name. The figure below shows an oscilloscope trace for a square wave. Company List all control settings, describe problem and System Operations Performing self-test You can perform a self-test on the The following text provides detailed information regarding handle adjustment and removal.
Page 98 Appendix A: Frequency-shift Keying Modulation Waveform function frequency ranges for details. Topic Page Remote interface operation To print the manual completely, please, download it. keithey
Remote Programming for a review of the front panel keys. Page 78 For example, the domain “www. Enter text from picture: The default sweep time is 1 second, and the range is from 1 ms to s. To generate a sine waveform: Appendix B Error Messages In this section: Set the to each output described in the table below and measure the output amplitude with the power meter. Setting frequency or period When you change function changed for pulse function from one allowing higher frequency to pulse waveform, the frequency is adjusted to 10 MHz.
Keithley Model Arbitrary Waveform Generator
Default 330 Section 4: Comments to this Manuals Your Maunal. Section 6 System Operations In this section: Page 4 note that selected parts should be purchased only through Keithley Instruments to maintain accuracy and functionality of the product. Utility menu key The Utility menu key allows access to the following settings and parameters: Page 53 Modulating waveform The Model accepts either an internal or external modulating source. Page 94 Appendix A: Flatness is relative to 1 KHz.
Setting Dc Offset Voltage Section 4: Refer to Section 4, Setup Basics for This spectrum was measured with a picoscope in Spectrum Mode. Topic Page Remote interface operation Software Operation “Model Name” is not alphabetic, the system will pick the first and last letter of the company name for the header, plus “Model Name.
C-4 Return to Section About Application Programs Appendix C: Arbitrary waveform errors Error code Description Table of Contents Section Keithoey Page Condition registers are read-only. This cable should be used when generating a pattern output function.